NIRSpec MOS Wavelength Ranges and Gaps
There is a physical gap between the JWST NIRSpec detectors, where MOS spectra of any resolution may lose wavelength coverage. High-resolution spectra may also have short- and long-wavelengths cut off at the outer edges of the detector. These effects depend on both the position of the shutter and the selected disperser.
Observations with the NIRSpec Micro-Shutter Assembly (MSA) can be affected by wavelength cutoffs due to the physical edges of the two detectors and the separation between them. How these cutoffs and gaps map to wavelengths depends on the shutter position and the disperser that was chosen for the observation. The full operating range for this instrument mode is from 0.6 to 5.3 µm. The nominal wavelength ranges for each disperser are given in NIRSpec Dispersers and Filters. Dithering during observation can help to recover missing wavelengths.
Visualizing wavelength cutoffs for observation plans
The JWST Exposure Time Calculator (ETC) can be used to show sample spectra (including cutoffs and wavelength gaps) from representative shutters across the MSA for each disperser-filter combination. For example, Figure 2 shows the ETC-produced spectra of one target in G235H/F170LP at three different MSA positions on quadrant 3. The spectra are overplotted on one another. Note the shift in the gap depending on shutter position.