Moving Target Calibration and Processing
The JWST calibration pipeline has the ability to co-add moving target exposures. Many steps in the pipeline are identical for moving and fixed targets; steps specific to moving targets are outlined on this page.
Main article: JWST Data Reduction Pipeline
JWST's science instrument calibration pipeline will support co-addition of moving target exposures. The calibration pipeline outputs products at 3 levels, with the unique steps for moving targets outlined below for each level.
Calibration pipeline level 1
Level 1 processing includes data formatting, science frame re-orientation, and calculation of WCS keyword values. For moving targets, the WCS is expanded to include 4 moving target keywords per integration and 2 additional keywords per observation. The first unique moving target keyword is created by triggering the moving target flag in APT:
MTWCS (moving target world coordinate system).
The right ascension and declination values of the reference pixel at the mid-time of the observation are recorded with the following keywords:
The right ascension and declination values of the reference pixel at the mid-time of each integration are recorded with the following keywords:
Then the following keywords are calculated using the standard WCS keywords
Dither and mosaic offsets are preserved.
Calibration pipeline level 2
Level 2 processing, or single exposure calibration processing, outputs calibrated single exposure count rate images. These steps are very similar to those for fixed targets, since the moving target is essentially "fixed" to the detector frame through telescope tracking. One unique consideration for moving targets is that the stars trail through the scene. The trails can cause transient increases in detected signal in the pixels they cross. In Cycle 1, these transient signals will be treated as cosmic rays in the ramps-to-slopes. In Cycle 2 or later a better treatment will be available.