JWST NIRCam imaging is obtained at 0.6–2.3 μm (0.031"/pix) and 2.4–5.0 μm (0.063"/pix) simultaneously over a 9.7 arcmin² field of view.
For standard imaging discussed here, NIRCam observes a 9.7 arcmin² field of view simultaneously in a short wavelength channel (0.6–2.3 µm) and long wavelength channel (2.4–5.0 µm) via a beam splitting dichroic. Extra-wide, wide, medium, and narrow filters are available.
When creating a NIRCam imaging observation with the Astronomer's Proposal Tool (APT), the primary options are the filter, dither pattern, subarray, and exposure time (via the detector readout patterns). One filter is selected for each wavelength channel. All other parameter selections apply to both channels.
The Exposure Time Calculator (ETC) allows users to determine the observing parameters required for their science goals.
Table 1. Properties of NIRCam short and long wavelength channels
|Short wavelength channel||Long wavelength channel|
|Wavelength range||0.6–2.3 μm||2.4–5.0 μm|
|Nyquist wavelength1||2.0 μm||4.0 μm|
|Fields of view||2 × 2.2′ × 2.2′ (with 4-5” gaps)||2 × 2.2′ × 2.2′|
|Imaging pixels||8 × 2040 × 2040 pixels||2 × 2040 × 2040 pixels|
1 PSF FWHM ~ 2 pixels; undersampled at shorter wavelengths.
Field of view
NIRCam's 2 modules operate in parallel to observe a 9.7 arcmin² field of view consisting of two 2.2' × 2.2' fields separated by a ~44" gap. Gaps of ~5" also separate the 4 detectors in each short wavelength module. If full-field imaging is not required, individual modules or smaller subarrays may be used to increase readout speeds and reduce data volumes. To cover larger areas, mosaics are best obtained in combination with primary dithers.
Twenty-nine NIRCam filters are available for imaging. They are classified as extra-wide (R ~ 1), wide (R ~ 4), medium (R ~ 10), and narrow (R ~ 100). One filter is selected for each wavelength channel, and both wavelengths are observed simultaneously.
Dithering and mosaics
Dithering is required for all NIRCam imaging observations (except time series monitoring). Dithering obtains multiple exposures slightly offset from one another on the sky. Primary dithers serve to fill image gaps, compensate for bad pixels, and mitigate flat-field uncertainties. Smaller secondary dithers provide subpixel sampling to improve the resolution of reduced images.
Mosaics are used to image regions of sky larger than the NIRCam field of view. A mosaic consists of a grid of N × M telescope pointings. Large NIRCam mosaics are best combined with a "FULL" primary dither pattern.
If the full NIRCam field of view is not required for science, smaller subarrays may be read out from the detectors. Faster readout times for subarrays allow brighter objects to be observed without saturation in shorter exposures. Subarrays also deliver lower data volumes than full detector readouts. Users interested in rapid cadence monitoring of bright, time-variable sources should refer to the time series imaging observing mode.
NIRCam's exposure time is governed by detector readout patterns.
Note that due to gaps between the detectors, dithered images will have uneven exposure times (with depth varying across the image).
The Exposure Time Calculator (ETC) allows users to determine the exposure time required to achieve their science goals.
The following sensitivity and saturation estimates are provided as a rough guide. Please use the Exposure Time Calculator (ETC) for your proposed observations.
NIRCam imaging is capable of studying very faint sources. Typical 10 ks images in F200W and F322W2 may yield S/N=10 detections of AB mag 29 (~9 nJy) point sources (and S/N=5 detections of AB mag 29.75 [4.5 nJy] sources).
Moderately bright sources (solar type, K-band Vega mag 14–17) will saturate the NIRCam detectors in full field broadband imaging. Narrower filters enable brighter saturation limits, as shown below. The smallest available subarray (64 × 64 pixels) brightens these limits by 6 magnitudes. The brightest objects are best studied using the NIRCam time series observing modes, which allow the use of weak lenses (at short wavelengths) and grisms (at long wavelengths) to spread the light over more pixels.
JWST User Documentation Home
Near Infrared Camera, NIRCAM
NIRCam Observing Modes
NIRCam Weak Lenses
NIRCam Coronagraphic Imaging
NIRCam Field of View
NIRCam GrismsNIRCam Imaging
NIRCam Detector Subarrays
NIRCam Detector Readout Patterns
NIRCam Time-Series Imaging
NIRCam Time Series Observations
NIRCam Subpixel Dithers
NIRCam Dithers and Mosaics
JWST Exposure Time Calculator, ETC
JWST Astronomers Proposal Tool, APT